- 05 Sep 2015 [14:17]
- ADJUST: +500ms/day to 40000 removed barometric compensator as an experiment to see how well the compensator has been working. In particular, I want to find out if the compensator also affects temperature compensation. For data this summer we seem to have dG/dp=-4ms/day per mbar and dG/dT= 46ms/day per deg [link] . This was worked out by finding the plane of best fit of the G(p,T) graph.